|Item Code||Item Name|
|ZMSP-C000||(Zircom Speed machine）|
|ZMSP-021216||High-Speed Firing Plate H|
|ZMSP-021218||High-Speed Firing Stter|
|ZMSP-021217||High-Speed Firing Tray|
NOTE: ZRCM-021215 Quick Firing Stand(H) is the same item Zircom Standard and Zircom Speed
NOTE: Zircom Speed has TWO type of Tray set. One for Zircom Standard ceramic tray set, other is for Zircom Speed carbon tray set.
NOTE: High Speed Firing Tray : Silicon carbide (sic)compound form, If zirconia material touches this tray or there is no High Speed Firing Plate, it will change the color (light YELLOW). Please do not contact the SiC tray with zirconia material. Always should be put on the High Speed Firing Plate. Also, It can not be a cause of this tray itself to GRAY colored zirconia materials. exception : reduction firing CO concentration up 4% or please attempt other company’s zirconia material.
The unit requires 220V single-phase, 20A and grounding
Unloaded Muffle Firing
When first purchasing the unit, please follow the unloaded muffle process listed below:
- Put proper amount of beads in the firing tray, then put one or two firing trays with its cover on.
- Choose the Course 0 and press [START STOP]. It takes approximately 7 hours.
How to Use:
- Powering on, then the unit displays Standby Mode
- Select the course by ▲▼
- Press [START/STOP]
How to Program:
Suppose you store a firing program on Program Course 5 as follows:
(Make sure firing table is down)
Course Name: TEST
Step 1 Temp: 1000C Time: 40min
Step 2 Temp: 1500C Time: 60min
Step 3 Temp: 1500C Time: 120min
Step 4 Temp: 400C Time: 60min
Step 5 Temp: 300C Time: 90min
Table Position: 5.
Start temperature is 20C. It is already programmed.
- Turn on the power to go to the Standby Mode
- Press ▲▼ button to choose the course number 5
- Press [F1]
- Press ▲ to move to Course Name
- Input TEST by using [F3]&[F4] and
- Prress ▼ to move back to Step 1
- Input 1000 by [F3]&[F4]
- Press to adjust time then imput 0:40 by [F3]&[F4]
- Press ▼ to move down to Step 2 then input 1500 and 1:00 by [F3]&[F4]
- Press ▼ to move down to Step 3 then input 1500 and 2:00 by [F3]&[F4]
- Press ▼ to move down to Step 4 then input 400 and 1:00 by [F3]&[F4]
- Press ▼ to move down to Step 5 then input 300 and 1:30 by [F3]&[F4]
- Press [F2] to change Setting Page (Table Position & Chme)
- Adjust Table Position to 5 by using [F3]&[F4]
- Press [F1] to return to Standby Mode
©Program will not be saved if you turn off the unit during programming.
- No4: Lifting Table Abnormal – Error occurs during lifting the firing
- No5: Heater Disconnection Abnormally – Heater is broken. Please replace it with a new heater
- No6: Thermocouple Error – The thermocouple is short-circulated or disconnected
- No8: Memory Error – The memory content is corrupted by the influence of noise or when the unit cannot read the memory
- No9: Excessive Rise of Temperature – The muffle temperature rises excessively
- No11: Top Cooling Fan Error – Error occurs in the top cooling fan
- No12: Under Cooling Fan Error – Error occurs in the lower cooling fan
For more information, please contact us.